Views: 0 Author: Site Editor Publish Time: 2023-03-28 Origin: Site
The bellows load cell can adopt two different input and output wiring modes:
1: One is four wire connection
The four wire load cell has no special requirements for the secondary instrument panel, which is very convenient for application, but when the cable is long, it is very easy to be affected by factors such as operating temperature fluctuations;
2: The other is six wire connection
The secondary instrument panel used together with the six wire weighing sensor has a feedback input socket. The application standard has certain limitations, but it is not easy to be affected by factors such as working temperature fluctuations. It has certain advantages in high-precision and accurate measurement and remote accurate measurement.